
Newest
Model of "PLACOM"
NSeries have made the excitingly highest attainement in the history of
Planimeters ever made.
Measuring capacity is remarkably enlarged.
Measurement by 6digit pulse count enables to measure 100 times larger
accumulative area than ordinary planimeters.
Easy calculations of Cumulative and Mean Value Measurement.
Conversion of Unit and Scale value is made by easy operations.
"WONDER", it is a most worthy word to be given to this "PLACOM"
NSeries. 
Special
Features of PLACOM NSeries
Measurement by 6digit
pulse
count
This NSeries measures the area by pulse count of 6digit. This method
enables a larger cumulative measurement of area (up to 10m²).
Maximum Cumulative
Measurement
Value of 10m².
With the measurement by 6digit pulse count, the NSeries Planimeter
can measure a larger area by 100 times than ordinary planimeter.
The conventional planimeter has a maximum cumulative area to measure
only 0.1 m². So, in measurement, at every measurement value of
0.1m², the display is to be reset to 0. This reqires to register
the numbers of how many times these resetting are occurred.
With the New PLACOM NSeries, there is no need of these troublesome
procedures.
Function of Easy
Conversion of
the Value of Unit and Scale.
IP900N and KP800N can convert easily the measured area value into a
new area value in a newly desired unit and with a newly desired scale
value.
KP92N and KP80N can convert the unit between in² and cm².
Hold Memory Function
In Cumulative Measurement, this Hold Memory function is effectively
used.
If an error occurs in measuring No. "N" th area, press C/AC key. Then
the pulse count value measured up to "N1" th appears. No need of
another measurements from the lst area to "N1"th area again.
Automatic shifting of
Unit to
upper Unit
When the measured area value overflows the displayed figures of 8
digits, the measured area value is calculated and displayed by an upper
unit. When the overflows is occurred in the uppermost unit, km²,
acre, the measurement can be continued by pulse count display.
Mean (Average) Value
Measurement.
In order to get better result of measurement, Mean Value Measurement is
recommended. This measurement can be easily done on this NSeries.
Measurement of an Area with different lateral and longitudinal scale.
DC
or AC power supply is available.
AutoPowerOff
function
for energy saving.
Specifications
of PLACOM NSeries
Model

KP90N

KP92N

KP80N

KP82N

Type

Roller
Type

Polar
Type

With
Computing Function

Single
Function

With
Computing Function

Single
Function

Display

Liquid
Crystal, 8 digit
Figures

10
Symbols

5
Symbols

10
Symbols

5
Symbols

Symbol

BattE,
SCALE, MEMO, HOLD
cm²,m²km²,in²,ft²,acre

BattE,
MEMO, HOLD
cm²,in²

BattE,
SCALE, MEMO, HOLD
cm²,m²km²,in²,ft²,acre

BattE,
MEMO, HOLD
cm²,in²

Converting
Function

Unit
and
Scale Value

Unit
(in²,cm²)

Unit
and
Scale Value

Unit
(in²,cm²)

Accumulated
Measuring
Value

Maximum
1m² ( scale
1:1)

Measuring
Range

•Maximum vartical
width:
325mm
•Horizontal roller rotating length: 30m

•Maximum,
with
the pole
put outside the pattern: 300Ømm
•Maximum, with the pole put inside the pattern: 800Ømm 
Accuracy

Within
± 0.2%
(within ± 2/1,000 pulses)

Power
Supply

1.
Builtin
NiCd Storage
Battery (rechargeable through AC Adapter
2. AC100V 120V, 220V and 240V, (using the supplied AC Adapter)
(Provided with auto poweroff function)

Operating
Duration

Approx.
30
hours of
continuous operation

Battery
Charging Time

Approx.
15
hours

Main
Unit
Weight

750g

750g

780g

780g

Accessories

Supplied
AC
Adapter and
Plastic Case

